While performing hydrogen gas release experimentation by thermally reacting a slurry of ammonia borane powder in silicone oil in a plug flow reactor, a discharge port on the test reactor became loose. A foaming white product was leaking from the fitting and discharging in the direction of the heat tape and insulation (back over the reactor). As a result, hot, reacting slurry flowed out of the port and was exposed to air. In the presence of oxygen, the slurry ignited, producing a green flame. A small green flame was noticed at the leak site and flaming product dripped onto the hood deck surface. The flame at the end of the reactor was ~10-12 inches tall at the highest point. The flame on the deck was ~4-6 inches in height.
The incident occurred behind the lowered sashes in the view more