A control room received a tank lower flammability limit (LFL) analyzer low sample flow alarm. The control room operator initiated the appropriate alarm response procedure and the facility entered limiting conditions of operation. At the time of the alarm, the facility was experiencing severe weather and the field operator was unable to investigate the alarm in the field. After the severe weather cleared, the field operator investigated the alarm and found the sample flow to be low and out-of-limits.
A hydrogen compressor had been shut down for repairs and was being put back into service when an explosion occurred, resulting in property damage. The compressor was equipped with interchangeable intake and outlet valves.
The discharge valve was installed in the intake valve position, causing the cylinder head to blow off and release H2 to the atmosphere. The ignition source was not indicated.
A gas mixture cylinder was connected to a Fourier Transform Infrared (FTIR) Spectrometer to purge residual carbon dioxide and water vapor. A staff member was preparing to use the FTIR instrument. Prior to use of the instrument, it must be purged with dry nitrogen to remove residual carbon dioxide and water vapor. When the gas mixture reached the instrument's globar (resistively heated ceramic) heat source, a localized explosion occurred. No injuries resulted from the explosion but the spectrometer housing was heavily damaged.