An incident involved an explosion of an oven that was heating decaborane for vaporization. In this incident, the heater controller was defective so the heating element was disconnected from the controller and plugged directly into a wall outlet. This situation allowed the oven to reach temperatures in excess of 400 °C within 20 minutes. While the temperature increased, the decaborane continued to expand, causing a significant pressure build-up within the oven. The pressure increase eventually caused the oven's viewing window to burst. A burst of burning hydrogen was emitted from the window and burned the face of a researcher who was hospitalized for approximately three weeks.
While attempting to replace a rupture disk in a liquid H2 vessel, H2 gas was released and ignited. In fighting the fire, liquid N2 was sprayed onto a second liquid H2 vessel located nearby. This resulted in cracking of the outer mild steel vacuum jacket. The loss of the vacuum caused a rapid increase in pressure and rupture of the burst disk of the second vessel. H2 boiled off and was burned in the fire.
The rupture disk was being replaced with a load of liquid H2 in the vessel and no separating inerting gas. The H2-air mixture was probably ignited by static discharges. Rupture of the second vessel burst disk was caused by the low-temperature exposure of the mild steel vacuum jacket.
A H2 air explosion occurred near a H2 compressor, located outside. Gaseous H2 had been released from a vent stack when a relief valve was actuated. The source of ignition was not known, but considerable damage was inflicted onto the system by the ensuing fire and explosion. Following the explosion, the shut-off valves were closed and the system was vented.
Two relief valves were located in the 3,000 psig system downstream of a 5,000/3,000 regulator. The relief valves were sized to handle substantially different flows. (One was designed for another program.) The relief valve was believed to have opened when the pressure setting was being increased from 2,700 to 2,900 psig. The accuracy of the 5,000 psig gauge used to control the dome of the 5 view more